The MX series is AES’ salt spray test chamber, and it is the only salt spray test chamber on the market with a 360-degree view into the chamber. Completely
The preparation of a salt solution consists normally of sodium chloride, containing 0.1% of sodium iodide and not more than 0.2% of total impurities, that is dissolved, 5 parts ion weight, into 95 parts of distilled water.
It is important to maintain this solution at a specific gravity of 1.023 to 1.037. The solution is maintained at a controlled temperature of 95°F by the test chamber's own integral temperature control system (temperature is adjustable).
Testing may be automated for up to 72 hours and even extended by adding an external storage tank, a tower level control and a salt spray solution reservoir level control to automatically refill the system.
Corrosion test chambers conform to military specifications including
The MX Series Salt Spray Test Chambers will accommodate the following specifications:
The MX Series Salt Spray Test Chambers will accommodate:
Standard 2 Independently controlled Watlow EZ-ZONE PM controllers for manual setting the tower and chamber
Dashboard view enables chamber control at the chamber, at a desktop, or remotely via laptop or tablet
Profile Builder has drag-and-drop, web-browser functionality
Data Logging with onboard, multi-year logging of live and current data, anyplace anytime
Communication and Control uses standard, well-documented JSON API for easy integration with other test platforms
Security decreases test interference with three authentication methods and six permission levels
Watlow F4T Programmable Controller
Features a 4.3” capacitive touchscreen minimizing the chance for error. Ethernet communication supported by AES|XCHANGE
100 OHM Platinum RTD is a highly-accurate sensor made from platinum components for increased sensitivity
We offer multiple types of safeties on our test chambers to help protect your product and your test chamber if anything malfunctions.
A heat fuse or thermal fuse, will interrupt the flow of electricity to the heater and shut it down if it reaches a certain temperature.